Structure of the PC measurement electronics with Spider8. B-1 .. When exchanging any modules, only original HBM parts must be used. Qualified personnel. Spider8 – Easy and reliable PC-based The Spider8 is a multi-channel PC measurement electronics for parallel, dynamic . Our factory in Darmstadt or HBM. Catman talks to Spider8 remotely. Keyword: Wireless. Remote communication with field equipment up to 5km away is now possible with HBM’s improved.
|Published (Last):||19 November 2009|
|PDF File Size:||8.94 Mb|
|ePub File Size:||18.38 Mb|
|Price:||Free* [*Free Regsitration Required]|
HBM Spider 8 Data Acquisition Unit
I solved the problems a few days ago. Furthermore, the system is very unstable. Can anybody help me?
I have 2 examples and one shows the mesured value in the port and the other plots graph in the screen of the signal mesured in the port. Spider with the P22 firmware version.
Trade In Your Spider8 For The New QuantumX! | HBM
If you need the archives email-me wanderleyff gmail. BUT the good days are gone…I have updated everything. Now I have got problems with data acquisition, it doesn’t work at all. I have updated everything. So I will give some remarks on what could have been the reason: First I chose a baud rate ofnow I use Perhaps the old firmware ubm some problems A big problem were the error messages which sometimes didn’t make sense to me. I guess, the best way is contacting a sales representative of HBM http: Best Regards Andreas S;ider8.
In most cases the baud rate will be the limiting factor, but don’t forget the measuring rate! Most Active Software Boards: There is one little disadvantage: I hope this help you. Message 7 of Reading values includes a lot of? I downloaded the LabView driver for this zpider8 device from the NI homepage.
Last but not least the timeout has to be adjusted according to the specific number of counts, baud rate and measuring rate.
I have run the firmware sp8firm32 and de archive hnm, both can be downloaded in the hbm website, and my example started to run. Message 3 of My OS is windows 98, labview version 6i. Any advice is welcome as well as was a place where I still can download the last firmware version. The Spider is connected with the notebook via the serial port. Message 5 of After some errors had occured, I didn’t turn the Spider off and on, and the error messages were strange.
So in this example, the timeout has to be adjusted in respect to the measuring rate. Message 1 of A further point are inconsistencies between the online help, the help function in the Spider setup program and the printed manual! It seems as the connection is not spidder8 enough. Message 10 of May be, he has a demo version for you almost catman professional including the catman activeX interface.
I was not able to find much in the hvm of documentation on the HBM website regarding the drivers you mentioned. Spider32 setup still can read it but spide8r a lot slower now. I haven’t downloaded the latest firmware yet. I cannot tell exactly why the data acquisition process didn’t work, but now everything is fine. This Tutorial has information regarding how to properly call functions from DLLs using the call library node. W henever this is not true, the Spider should be turned off and on.
First of all, my operating system is NT4.
Lot 753: HBM Spider 8 Data Acquisition Panels
First at all say that my spider8 is version P13 quite old. Message 4 of This is not very useful! HBM home page doesn?
Message 2 of